ESD - Generators verified according to the standard International Electro technical Commission 61000-4-2.
The measurements of the single pulses take place with a special 2 ohm shunt. For the correct verification of the ESD impulses an oscilloscope with a minimum band width of 1 GHz is necessary.
The ESD verification set SESD 8800-4 works to 4 GHz and is appropriate for the future normative measurements with a band width of 2 GHz (see newest draft standards)
2 Ohm ESD – target (4 GHz), 20 dB attenuator, Koax cable 1m, Calibration certificate (PTB) with data of the S-parameters in the frequency range up to 4 GHz